for standard waveforms. The Agilent Technologies A Use the A to generate complex . Manual language options (please specify one). ABA US. For Safety information, Warranties, and Regulatory information, see the last page in this manual. Agilent A. Option Phase-Lock Assembly. oscilloscope or DMM and send it to your HP A for output. 3-year warranty. With your HP A, you get operating and service manuals, a quick reference .
|Published (Last):||16 January 2005|
|PDF File Size:||7.79 Mb|
|ePub File Size:||18.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
Index front panel annunciators, 5 factory setting gate source dc volts selection, 24 baud rate RS, burst modulation, downloading waveforms, – calibration security code, gated burst mode, 81, menu examples, 33 – 39 data bits RS, GPIB 31320a messages, This is a used manual so there may be staining,tears and other damage from use in a working environment.
Inthe amplitude of the carrier is varied by the amplitude of the modulating waveform. Inthe frequency of the carrier is varied by the frequency of the modulating waveform.
The carrier output can be controlled using either triggered or externally- gated methods.
Agilent Technologies 33120A User Manual
You can set the offset to a positive or negative number with the restrictions shown below. All of the standard output functions except dc and noise have an associated sync signal.
Mouse over to Zoom – Click to enlarge. The driver is useful, thank you. The default amplitude is mVpp into 50 ohms for all functions.
Please enter a number less than or equal to 1. You can output the arbitrary waveform directly from volatile memory as described in step 2 or you can copy the waveform to non-volatile memory. Seller assumes all responsibility for this listing.
At power-on, the units for output amplitude are volts peak-to-peak. Chapter 2 Front-Panel Menu Operation A front-panel menu tutorial Menu Example 1 The following steps show you how to turn on the menu, move up and down between levels, move across the choices on each level, and turn off the menu. Page Index string calibration, table of contents, 9 – 11 units output amplitude display message, technical specifications, – front-panel selection, 64 swapped byte order, terminals remote interface selection, SWE: Any international shipping and import charges are paid in part to Pitney Bowes Inc.
Select a Web Site
This amount is subject to change until you make payment. Instructions and mounting hardware are included with each rack-mounting kit. Freq 1 Enable the frequency modify mode. Subject 3312a0 credit approval. If you have questions relating to the operation of the A, call in the United States, or contact your nearest Agilent Technologies Sales Office. Agilent a output Disable.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
Thanks for all your reply. Set the carrier frequency between 10 mHz and 5 MHz kHz for triangle and ramp. Check to see if you have enclosed the character string in single or double quotes and that the string contains valid characters.
Page Error Messages The five built-in arbitrary waveforms can be output at a maximum of 5 MHz. The default is Sine.
Agilent a output Disable – Discussion Forums – National Instruments
Thanks for the reply. The commands used for burst modulation are listed on page Seller information galacticmarketplace Select a Web Site Choose a web site to get translated content where available and see local events and offers.
You can use sine, square, ramp, triangle, or arbitrary waveforms for the hop frequency waveshape.
Page Index points, number in arb waveform, RS shipping inventory, 15 ports, serial COMarb download example, short form, SCPI commands, power-down recall,baud rate,sinc waveform, power-line voltage cables and adapters, single triggering, fuse, 16 configuration, However, you cannot use the noise function agipent dc volts as the carrier waveform. Recall 5 Recall the stored instrument state.