The Boat Anchor Manual Archive Path: /hp/a/. This is the BAMA archive. These manuals are available for download and free of charge. I am now accepting. View and Download HP A operating and service manual online. DIGITAL VOLTMETER. A Measuring Instruments pdf manual download. Hp A Pdf User Manuals. View online or download Hp A Operating And Service Manual.
|Published (Last):||23 October 2008|
|PDF File Size:||7.96 Mb|
|ePub File Size:||20.61 Mb|
|Price:||Free* [*Free Regsitration Required]|
The error mode of the math feature is described by the formula: The transfer time depends on the num- ber of ASCII character send to the instrument and the response lime of the controller. General Description Set the Test Oscillator frequency to 10 kHz and adjust its level controls for the 5 V A reading recorded in step m. In any correspondence, identify the instrument by up number and full serial number.
It calls attention to a pro- cedure, practice, condition or the like, which, if not correctly per- formed or adhered to. Select a valid country. The A reading should be between 9.
manua Record the A reading: Basic Ohms Measurement Diagram. Here, the DVM appears to be in tolerance 0. The A reading should be within 0.
Agilent / HP 3455A Digital Voltmeter
Alternating current power line. Inguard Section From Figurethe following observa- tions can be made: Derive 6-month test limits fromOhms Accuracy specifications listed in Table If a SA input voltage greater than V is needed, the following procedure should always be followed.
Do not apply negative voltages greater than V dc. Return the instrument to a Hewlett- Packard Sales and Service Office for service and repair to ensure that safety features are main- tained.
The inverter operates at a frequency of The resultant positive signal is applied to the input of the squaring amplifier. The value of the unknown resistance is computed by the main controller. Input resistance of the AC Converter is approximately 2 3455z. Code List of Manufacturers.
Proudly powered by WordPress.
For example, if the calibration chart indicates that the A output is 0. Exchange Assemblies 6- Adjust R72 10 V.
| HP B – Calibration module from HP A
If the appropriate power cable is not included with the instru- ment, notify the nearest -hp- Sales and Service Office and the proper cable will be provided.
If higher accuracy is desired, an ac-to-dc thermal transfer techni- que Figure can be used. Learn More – opens in a new window or tab Any international shipping and import charges are paid in part to Pitney Bowes Inc. See Table for list of manufacturers.
The A reading should be 0.
Hpp self test will be repeated until another function is selected. Function and location of the part. The following is an example of a data message output by the A: